Measuring Distances in Imaris’ Surpass View
|Within the MeasurementPro module Imaris offers tools for making
quick, manual distance measurements directly on your 3D image. By using Measurement Points you can easily obtain the distance between two structures or
between arbitrary locations by interactively specifying those positions simply using mouse clicks. Measurement points can be placed anywhere within your 3D
volume data; either on Surpass objects (Spots, Surfaces, Cell, Filaments, Ortho and Oblique Slicers) or in the volume rendered image to measure distances
amongst non-segmented structures of interest.
You could imagine that placing points at the correct depth within a 3D image is a cumbersome task. However, Imaris uses an AutoDepth algorithm to
automatically position the points at the correct depth within the 3D image. Depending on your selection the placement of a point is determined by either: 1)
the most intense voxel of all channels directly underneath the cursor, 2) the most intense voxel of a specific channel directly underneath the cursor or 3)
previously created solid objects. (Surface, Spot, Filament and Cell objects as well as Ortho and Oblique Slicers are all considered “solid objects”.)
The first two options are applicable if you wish to insert Measurement Points in a volume rendered data set. When measuring distances from, for example,
a Surface to a Filament then the solid Objects option is appropriate. It is also possible to measure distances between solid objects and points of interest
within (non-segmented) volume rendered data.
Start a new set of measurements by adding a Measurement Points object to your current Surpass Scene .
While on the Edit tab (Fig 1) and in Select mode, add a new point by holding Shift and clicking the left mouse button at the location of your choice
within the 3D image. The new point will either be placed at the correct depth determined by either voxel intensities or by the presence of a solid object
underneath the cursor as described above. After placing a second point a new line connecting the two points is created. Distances can be calculated 2
points at a time (Lines; Fig 2) or by consecutively connecting all points (Polygon; Fig 3). This option is found on the Measurement Points Settings
For easy editing, points can be moved after placement. The point must first be selected (left mouse click) and then the correct numerical values can be
entered into the X, Y, Z fields.A point can also be moved by first selecting it and then using Shift + left mouse click to place it in another position. All
of these features are found on the Edit tab, where you can also adjust the name of the point or delete it. Deleting all points is also an option found on
the Edit Tab.
To change how points are displayed in the 3D Surpass view and to customize the display of the information provided for line measurements, use the
options found on the Settings tab (Fig 5).
Here, you can choose from three different point shapes (tetrahedron, cube, or sphere; Fig 4). You can also change the line width and color. Options on
the Color tab allow you to assign a different color for each unconnected line.
Once a line is created, Imaris displays the intensity of each voxel along the line. To display the Intensity Profiles along the lines click on the
Histogram tab (Fig 6).
The x-axis of the histogram represents the length of the line, while on the left-hand side of the histogram, the channel(s) intensity range is
displayed. Below the histogram, you can find the channel and line selection. Here you select the channels and lines that should be displayed. The histogram
can display any or all measurement pairs. Clicking on the Snapshot button creates a still image (.tiff) of the histogram and intensity range(s).
Under the Statistical tab (Fig 7), the results of each line measurement are shown.
You may be wondering how to place a point within a specific plane of the image when it lacks a solid object or high intensity voxel in that particular
Ortho and Oblique slicers are ideally suited for this occasion because they can be placed at a specific plane and are solid Objects (Fig 8).
First, insert a new Slicer and position it at a plane where you wish to place a new Measurement Point. Then, switch to a Measurement Points object, select
solid Objects and place a new point on the Slicer.
If you have questions about the features mentioned here or any other features found within Imaris please contact your local Sales or Support person.